Volume 22 Preprint 34
Experimental study on accelerated corrosion technique of OPC and PPC beams in coastal environment
Keywords: Accelerated test, Corrosion rate, Load-Deflection curve, Linear polarization resistance
In this paper corroded Ordinary Portland Cement (OPC) and Portland Pozzolana Cement (PPC) beams were tested in lab to evaluate flexural behaviour of beams. Accelerated corrosion technique was adopted to corrode the beams up to 10% experimentally. Corrosion was measured using Applied Corrosion Monitoring (ACM) instrument. It is observed that PPC beams were more corrosion- resistant than OPC beams. As rate of corrosion increases, the deflection also increases but load carrying capacities of the beams were reduced, for both OPC and PPC beams. It is concluded that the stiffness of the beams is reduced nonlinearly when rate of corrosion is increased.
Because you are not logged-in to the journal, it is now our policy to display a 'text-only' version of the preprint. This version is obtained by extracting the text from the PDF or HTML file, and it is not guaranteed that the text will be a true image of the text of the paper. The text-only version is intended to act as a reference for search engines when they index the site, and it is not designed to be read by humans!
If you wish to view the human-readable version of the preprint, then please Register (if you have not already done so) and Login. Registration is completely free.
Review of “Application of the Electrochemistry Noise Technology to the
Evaluation of the Corrosion Inhibitor” by Jun Cui, Yuansheng Pei
While this paper on the application of EN techniques to the study of inhibitor performance is
potentially useful, the paper suffers from a number of weaknesses:
The English is poor, such that it is often unclear and in at least one case says the opposite of
what I believe is intended.
The description of experimental and data analysis procedures is inadequate.
There are questions about the quality of the measured data and the accuracy of some of the
It may be possible to produce a revised paper that is suitable for publication, but only if it can be
demonstrated that the measured data are valid.
L44 “However, the weight loss experiment consumes a large amount of materials owing to the
non-reusable of the metal substrate;” unless the material is extraordinarily expensive this is a very
minor problem compared to the other costs involved.
“in-suit” presumably “in-situ” is intended.
L65 “However, rare studs apply EN technology to evaluate the passivation behavior of the
corrosion inhibitor.” - while there may be relatively few studies using EN on inhibited systems,
there have been some, and these should be mentioned.
“a controlled-release inhibitor” this is an inadequate description of the inhibitor chemistry.
L100 “44% of epoxy value” what does this mean?
L102 “polisher powder” inadequate description.
L109 I believe the Autolab potentiostat does not have anti-aliasing filters that can be adjusted to
be appropriate for the sampling frequency in use, so there is a significant risk that the measured data
are contaminated by aliasing. The validity of the measured EN needs to be confirmed (e.g. by
following the procedure in “Guideline for an assessment of electrochemical noise measurement
devices”, S. Ritter, F. Huet and R.A. Cottis, Materials and Corrosion-Werkstoffe Und Korrosion,
Volume: 63, Issue: 4, Pages: 297-302, 2012).
L116 “five-order fitting method” presumably this really means subtraction of a fifth order
polynomial, but the significance of this remains unclear, since there is no mention of the time record
duration to which this trend removal was applied.
L126 More information needed on EIS parameters, particularly number of frequencies per decade
and overall measurement time.
L146 “an acceptable passivation performance was obtained until the total boron concentration
over 100 mg L-1” should this actually say “… was not obtained ...”?
L147 It appears that the objective of the test procedure on the controlled release inhibitor was to
test in reasonably constant inhibitor concentration, in which case why was a controlled release
L285 30 days is hardly ‘long-term’ in a real application.
L395 “shot noise”.
The normalization of the current and potential signals with respect to area makes no
sense at all. For absolute values of current and potential it is reasonable to report current per unit
area and potential independent of area, but for EN the effect of area is less clear, so it is best just to
report coupling current and potential (see “The Interpretation of Electrochemical Noise Data”,
R.A.Cottis, Corrosion, 27, 3, 265-285 (2001)). It is also unclear what is actually being plotted – are
they the ‘raw’ potential and current (this seems unlikely as the signals display little short-term
variation) or the results after trend removal or some other statistical treatment such as standard
Figures 4, 9, 13
Why is 1/Rn plotted in 4 and Rn or Rp in 9 and 13 – this is just confusing?
Also the 1/Rn values in 4 seem completely incompatible with the Rn values in 9 and the Rp values